Payment & Shipping Terms:
|Power Supply:||DC24v||Max Connect:||8Loadcells|
|Open Swith Input:||3DI||Open Repay Output:||4DO|
|Protection Level:||IP65||Static Weighing Accuracy:||0.2%~0.5%|
electronic weight indicator,
digital load controller
EMC design with high anti-jam for industrial environment.
6 LED/VFD digital tubes for English&digit display.
64-segment LED for percent display.
24-bit ∑-△ADC with internal resolution 1/1,000,000.
High sampling frequency 200Hz.
Special Anti-vibration Digital Filtering Algorithm for precise weighing, stable display and rapid response.
Auto Zero Tracking.
Load Calibration &, Data Calibration.
The Weighing Record can be printed.
Definable AO/COM[Communication Port].
Power Supply: DC24V±15%, 50/60Hz, Max. 10W.
Loadcell Excitation Voltage/Current: DC10V/250mA.
8 Loadcells[350Ω] connectable.
Weighing Signal Input Range: 0~25mV.
3 Normally Open Switch Inputs [DI].
4 Normally Open Relay Outputs [DO]: AC250V/DC24V, 1A.
1 Analog Signal Output [AO]: 4~20mA, 0.05%FS.
Accuracy Grade: III.
COM1: Fixed RS232&RS485. Free to use one of them.
COM2: Optional RS232/RS485/RS422/Profibus-DP/Ethernet.
Connect Host IPC, Remote Display, Printer&Wireless Module.
Outline Size [W×H×D]: 160 × 84 × 188 mm.
Panel Cut-out Size [W×H]: 151 × 76 mm.
Operating Temperature: -25℃~+45℃.
Protection Level of Front Panel: IP65.
Verification Accuracy: 0.02%.
Static Weighing Accuracy: 0.2%~0.5%.
The warranty period for our products is 12 (twelve) months.
If the products from SUPMETER are proved to be unable to work normally due to unartificial reasons, and the problems can't be solved by customers under the technical support from SUPMETER, then customers can send the problematic items back to SUPMETER. SUPMETER is RESPONSIBLE to MEND (or REPLACE when necessary) them FREE OF CHARGE.
(Note: if after checking, the returned goods are proved to be in good condition, the trouble is resulted to be caused by improper operation by customers, SUPMETER is IRRESPONSIBLE to bear the shipping & other charges related to the returning.)
Contact Person: He